This commit does not belong to any branch on this repository, and may belong to a fork outside of the repository.
Failed to load latest commit information.
TH260Lib Programming Library for TimeHarp 260 Version 22.214.171.124 PicoQuant GmbH - February 2020 Introduction The TimeHarp 260 PCIe board is a Time Correlated Single Photon Counting System for applications such as time resolved fluorescence measurement. The system requires at least a dual core computer with at least 2 GB of memory and 1.5 GHz CPU clock (2 GHz and 4 GB recommended). The TimeHarp software is suitable for Windows 8.1 and 10. The programming library is a DLL with demos for various programming languages. Please refer to the manual (PDF) for instructions. Note that you must purchase the TimeHarp 260 DLL option for this software to work. This is a one time fee, maintenance updates are free. What's new in version 3.2 - Corrects an issue of small time shifts (a few hundred ps) on the histogram time axis, occurring from one hardware initialization to the next - Fixes an inversion of the selected input edge (TimeHarp 260 NANO only) - Fixes an issue with warnings occasionally being incorrect at the time of first retrieval after initialization - Provides some minor documentation updates - API and data structures remain unchanged - The set of demos has been extended to inlcude Python and the LabVIEW demos have been significantly refurbished What's new in version 126.96.36.199 - Fixes an issue with sporadic timeout errors upon hardware initialization - Fixes an issue where errors occurred when Windows was set to 1 or 2 ms timer resolution - Provides a new driver to support Windows 10 with “secure boot” enabled - Fixes some minor documentation errors - API and data structures remain unchanged What was new in version 188.8.131.52 - Fixes an issue with stopping measurements on some PCs - Improves histogramming throughput while reducing CPU load - The API and data structures remain unchanged What was new in version 184.108.40.206 - Supports new hardware models manufactured after February 2017 - Fixes a buffer alignment issue causing freezes on some recent PC mainboard models - Fixes an issue with hardware debug information retrieval - Improves data throughput in histogramming mode - The API and data structures remain unchanged in functionaity - For best throughput observe new buffer alignment recommendation What was new in version 220.127.116.11 - Fixes an issue where unfortunate calling sequences could lead to the loss of status information retrieved via TH260_GetFlags - Some documentation improvements What was new in version 3.0 - Supports the latest hardware improvement of the TimeHarp 260 N - now running at 250 ps resolution (owners of older models with 1 ns resolution can request a quote for a hardware upgrade to 250 ps) - Officially supports Windows 10 - Fixes some minor bugs - API and data formats remain unchanged What was new in version 2.0 - A new library routine SetInputDeadTime for suppression of some detector artefacts. (Note that this works only for TimeHarp 260 P purchased after April 2015. Old boards can be updated but must be returned to PicoQuant for this purpose.) - A bugfix in the shutdown code called upon unloading the library - Some small demo code improvements - A new device driver with improved error handling and automated installation. The driver must meet new code signing requirements imposed by Microsoft. Consequently Windows versions prior to 7 are no longer supported. If you must use an older Windows version you can still use the driver of the TimeHarp 260 release version 1.1. - Some documentation fixes Disclaimer PicoQuant GmbH disclaims all warranties with regard to this software and associated documentation including all implied warranties of merchantability and fitness. In no case shall PicoQuant GmbH be liable for any direct, indirect or consequential damages or any material or immaterial damages whatsoever resulting from loss of data, time or profits arising from use or performance of this software. License and Copyright Notice With the TimeHarp 260 DLL option you have purchased a license to use the TH260Lib software. You have not purchased the software itself. The software is protected by copyright and intellectual property laws. You may not distribute the software to third parties or reverse engineer, decompile or disassemble the software or part thereof. You may use and modify demo code to create your own software. Original or modified demo code may be re-distributed, provided that the original disclaimer and copyright notes are not removed from it. Copyright of the manual and on-line documentation belongs to PicoQuant GmbH. No parts of it may be reproduced, translated or transferred to third parties without written permission of PicoQuant GmbH. Trademark Disclaimer HydraHarp, PicoHarp, TimeHarp and NanoHarp are registered trademarks of PicoQuant GmbH. Other products and corporate names appearing in the product manuals or in the online documentation may or may not be registered trademarks or copyrights of their respective owners. They are used only for identification or explanation and to the owner’s benefit, without intent to infringe. Installation Before installation, make sure to backup any work you kept in previous installation directories and uninstall any older versions of TH260Lib. The TH260Lib package can be distributed on CD/DVD or via download. The setup distribution file is setup.exe. If you received the package via download it will be packed in a zip-file. Unzip that file and place the distribution setup file in a temporary disk folder. Start the installation by running setup.exe and follow the installer wizard. The setup program will install the programming library including driver, manual, and programming demos. Note that the demos create output files and must have write access in the folder where you run them. This may not be the case in the default installation folder. If need be, please adjust the acces permissions or copy the demos to a more suitable folder. Before uninstalling the TH260Lib package, please backup your measurement data and custom programs. From the start menu select: PicoQuant - TimeHarp 260 - TH260Lib vx.x > uninstall. Alternatively you can use the Control Panel Wizard 'Add/Remove Programs' (in some Windows versions this Wizard is called 'Software') Contact and Support PicoQuant GmbH Rudower Chaussee 29 12489 Berlin, Germany Phone +49 30 1208820-0 Fax +49 30 1208820-90 email firstname.lastname@example.org www http://www.picoquant.com