What is it?
OpenMind is a take on an open source electroencephalograph, i.e. a device to capture "brain waves" - the tiny electrical signals your brain emits when working. By itself, the hardware can also be used as "just" a pretty high-resolution albeit slow ADC to upgrade your arduino (imagine gaining 12 bits of resolution...). I suspect that the hardware should also easily be sensible enough to directly measure muscle action potentials.
This repository contains the third version of the hardware. This is a modular system: Each PCB can (depending on what chip you put on it and what you want to do) measure signals on one or two channels with one (absolute) or two (differential) electrodes each. The chips used are ADS129X, 24-bit ADCs with integrated AC hum filter and preamplifier with differential inputs. The ADS1291 contains one, the ADS1292 two channels. In a single-channel configuration one board is the digital equivalent of a conventional EEG's active electrode.
All these "digital active electrodes" can be connected on one SPI bus with one !CS-line per module - so a bunch of eight of them can be controlled via three SPI pins and 8 !CS pins (which themselves can be multiplexed from three pins using a small logic IC)
2ch-dual-shieldrecommended A 2-channel Arduino shield which may be populated as an analog frontend for the Arduino's ADCs or (recommended) with an integrated analog frontend containing a 16- or 24-bit ADC.
4ch-1chip-minimal-breakoutA minimalist breakout board for a large (TQFP-64) 4-channel chip. The coax electrode cables are supposed to be soldered directly to the board.
4ch-dual-shieldA variant containing 4 differential channels which can be populated analog for use with the Arduino's ADC or with integrated TI frontends. This board became way too crowded, I recommend its successor, the 2-channel version.
active-electrodeAn active electrode design small enough so the board can be the electrode. In this design, the analog frontend sits on the electrode so there are no analog connections leaving the board whatsoever (except for the analog power supply, that is).
analog-shieldA fully analog Arduino shield containing nil but 4 primitive differential amplifiers to connect 8 inputs to 4 adc channels of the Arduino.
libraries-and-stuffLibraries used in the boards. As of now, not all required libs are included but just the ones I made myself. For the others I still have to sort out the licensing foo.
In absolute mode you connect one electrode to each channel's positive (non-inverting) input, and connect the negative (inverting) input to the reference signal (as denoted on the schematic/PCB). In differential mode you connect one electrode to each of the two inputs of each channel and the device will measure the potential between each pair.
Right-leg drive (RLD)
The RLD is an op-amp which is used to cancel out LF interference with the EEG signal. It works by continuously comparing
a reference potential (normally VDD/2) with an average of all electrode signals and driving a dedicated reference
electrode with the difference. This signal is very weak and thus not noticable except in the measurements.
Multiple devices can be cascaded by connecting their
RLD_INV pins together and powering down all but one RLD. This
RLD's output signal is used to provide the reference voltage.
The active RLD can be selected at runtime, and you could even select the reference electrode at runtime from any
electrode connected to one of the inputs - sacrificing that one electrode.
The ADS12XX does contain a pretty neat lead-off-detection, which can be used to measure the connection of the electrode to the skin and the connection of the RLD electrode to the skin. This could e.g. be used to signify the electrodes connection status with the on-board LEDs.
RLDIN/RLDREF signal can be routed to any of the device's electrodes (whatever that might be needed for) and/or
any of the ADC inputs - so you can connect some auxiliary signal to this pin and measure it with any ADC.
The pad on the bottom side of the board is intended to be used to surface-mount type 261k or 269k 9v battery connectors (available at digikey). I did not yet physically check it, but I will as soon as I order at digikey.
To guarantee galvanic separation from mains potential and to improve the power supply noise floor for the ADCs I would recommend using a battery (rechargeable or not), radioisotope thermoelectric (or other peltier), fuel cell or other non-switching, off-grid power supply. Extra style points are given for the radioactive option.
The interfacing of these ADCs is pretty straightforward. A driver for AVRs currently only supporting the ADS1194 (16-bit, 4ch version) can be found at https://github.com/jaseg/OpenMind-firmware
What to do with this?
Just a bunch of ideas:
- http://www.instructables.com/id/Animatronic-Cat-Ears/ This only needs more pony.
- A sleep cycle-detector that wakes you up by using some LEDs to simulate dawn at exactly the right time
- A muscle-computer-interface, i.e. a device that senses when you physically move your muscles by measuring the pretty strong signals your brain sends to them. For reference see William Gibson: Burning Chrome (though, in comparision to the one used there, openmind is still one-way).
These schematics and code are distributed in the hope that they will be useful, but WITHOUT ANY WARRANTY; without even the implied warranty of MERCHANTABILITY or FITNESS FOR A PARTICULAR PURPOSE. Now, the following should be obvious: Do not connect this stuff to anything alive unless you absolutely know what you are doing and do not blame me when you fry your brains with it.