DSO-138 is an excellent piece of hardware based on ARM Cortex M3 core STM32F103 processor and sufficient for most beginner users. The stock firmware, while quite responsive, can use a few improvements. The main shortcoming which prompted the development of DLO-138 firmware is the inability to get waveform data into a computer for further analysis and the lack of a second channel. Engineers troubleshooting hardware issues need to mark reference points on waveform so having another analog or digital channel can greatly improve analysis. This firmware hopes to improve on these issues.
- Two analog channels
- Two digital logic channels (SWDIO and SWDIO pins (PA13 and PA14) on board)
- Serial port interface for captured waveform data
- Trigger source selectable from Analog Channel 1 or Digital Channel
- Option to use rotary encoder instead of + - and SEL switches
- 2K sample depth
This firmware can be used on stock DSO-138 hardware as well. Select one of the pre-compiled binaries to suit the board. Follow the firmware upgrade instructions for DSO-138. At any time, you can reflash DSO-138 with JYE Tech provided firmware.
Extra features come at an additional cost. In the case of DLO-138, it is the loss of lowest timebase. Maximum sampling rate in DLO-138 is 20 µs/div instead of 10 µs/div. In the 20 µs/div range, firmware under-samples ADC channels, often reading same data twice. To use the second analog channel, analog front end has to be duplicated on a daughter board. On a stock hardware, this firmware can be used to provide two digital logic channels.
The build environment uses Arduino. For help with setting up IDE visit http://www.stm32duino.com
Push button in encoder (SEL if using switches) moves focus to next parameter Left/Right turn in encoder (+/- if using switches) changes the parameter which is in focus Short press OK to HOLD the waveform and output it on serial port Long press OK button: Focus Action Trigger Level Zero the trigger level to Analog channel 1 Wave X scrollbar Center waveform on screen (at trigger point) Wave Y cursor Zero the cursor. If Analog CH1 coupling is GND, waveform reference base is set Other Toggle on screen Analog CH1 statistics display Press and hold OK button at power up to reset settings to default
STM32Duino - http://www.stm32duino.com
Adafruint Graphics Library - https://github.com/adafruit/Adafruit-GFX-Library
Parallel 8 bit ILI9341 library - https://github.com/stevstrong/Adafruit_TFTLCD_8bit_STM32